Specification and model:
Continuous CBI, continuous + automatic CBI + PVC
Continuous infusion (CBI) type is powered by the mechanical elasticity of the drug storage bag, which is limited to the flow to the patient through the microporous tube to implement safe, effective, micro and continuous uniform injection of the drug solution to achieve the analgesic effect; the squeezing pump body does not affect the normal flow rate of the drug solution; the drug solution filtering device can better control the particles entering the human body.
CBI + PCA is powered by the mechanical elasticity of the drug storage bag, and it can be injected safely, effectively, in a small amount and uniformly to the patients through the microporous tube to achieve the analgesic effect;
Automatic dosing function, that is, the patient can add medicine according to the pain situation. The squeezing pump body does not affect the normal flow rate of the liquid medicine; the liquid medicine filtering device can better control the particles entering the human body.
Scope of application:
In clinic, the main analgesic measures are: postoperative, cancer, painless delivery, angina pectoris and other causes of acute and chronic pain. It is beneficial to wound healing; reduce complications; and other adverse reactions caused by pain.
1、 After inspection, if there is no damage and the opening package is in good condition, tear the package bag along the opening and take out the infusion pump, check and confirm whether all parts are connected tightly and firmly.
2、 Open the protective cap of the dosing port, clamp the stop clip, unscrew the protective cap of the three-way valve, pump the prepared liquid medicine into the sterile syringe (without needle) and inject it into the pump body through the three-way valve.
3、 After dosing, if there is any gas: open the stop clip, turn the pump upside down, do not remove the handle pressure strip, and discharge the gas until there is no bubble in the pipe. At the same time, observe the outflow of the liquid medicine. It takes about 25 minutes. After clamping, the current stop clamp is ready for use.
4、 PCA key is the function key of adding dosage automatically by the patient on the basis of calibrating the normal and continuous dosage per hour. This operation must be carried out under the guidance of medical staff, and fill in the corresponding treatment card, so as to understand and analyze the patient's condition in the process of PCA additional times. To achieve the best effect of analgesia.